Glove Box Day-to-day Upkeep Guideline
one) Fill in the use history in time following functioning the glove box, which include "user, drinking water oxygen benefit, what to do"
2) When changing fuel cylinders in the glove box, the stress is established to the default value ± 3mbar
three) The every day functioning tension is set according to the operating ease and comfort. When not in use for a extended time, the stress is established to one-6mbar
4) Consistently check the don of the gloves, check the oil amount of the vacuum pump (in between the upstream and downstream markings), and include the pump oil properly
five) Clean the drinking water probe once every 2 to 3 months to make certain precise detection. (Also want to refer to the true predicament of the natural environment in distinct consumer boxes
six) If you usually use organic solvents in the glove box, it is encouraged that you consistently change the organic and natural solvent adsorbent substance, not later than 1 yr at the most up-to-date
seven) Change the filter element frequently, no later than two decades
8) In accordance to the true use of your vac glove box box, come to a decision to substitute the vacuum pump oil cycle, this sort of as black and yellow pump oil, water or extra natural and organic solvents in the pump oil. The oil substitute cycle really should not exceed 2 a long time at the most recent
9) If a significant sum of organic and natural solvent is inevitably uncovered during the experiment, the cycle requires to be stopped for the experiment. Just after the experiment, initial thoroughly clean and then start out the cycle
10) When the setting in the box is poisonous or unsafe gasoline, the exhaust port of the vacuum pump, the exhaust port of the regeneration fuel and the exhaust port of the cleansing valve should really be exhausted to the out of doors with an exhaust pipe (when exhaust is not authorized to be outside, the exhaust gas remedy need to be completed to start with).